A pocket refract meter design on the principle of total reflection for measuring the refractive index. Built in ocular scale, on inspection, gives direct reading for percentage of sugar density.
A. Range 0-32% in division of 0.2%
B. Range 28-62% in division of 0.2%
C. Range 58-92% in division of 0.2%